|View Full Text||PubReader|
|Abstract||Print this Article|
|PMC||Export to Citation|
|Article as PDF||Open Access|
Exp Neurobiol 2009; 18(2): 137-145
Published online December 31, 2009
© The Korean Society for Brain and Neural Sciences
Hyun Joo Lee1, Selenge Nyamdorj2, Hyung-Cheul Shin1† and Jae Mok Ahn2*
1Department of Physiology, College of Medicine, 2Department of Electronics Engineering, College of Information & Electronics Engineering, Hallym University, Chuncheon 200-702, Korea
Correspondence to: *To whom correspondence should be addressed.
†These authors contributed equally to this work.
TEL: 82-33-248-2347, FAX: 82-33-241-4183
A brain-machine interface (BMI) has recently been introduced to research a reliable control of machine from the brain information processing through single neural spikes in motor brain areas for paralyzed individuals. Small, wireless, and implantable BMI system should be developed to decode movement information for classifications of neural activities in the brain. In this paper, we have developed a totally implantable wireless neural signal transmission system (TiWiNets) combined with advanced digital signal processing capable of implementing a high performance BMI system. It consisted of a preamplifier with only 2 operational amplifiers (op-amps) for each channel, wireless bluetooth module (BM), a Labview-based monitor program, and 16 bit-RISC microcontroller. Digital finite impulse response (FIR) band-pass filter based on windowed sinc method was designed to transmit neural signals corresponding to the frequency range of 400 Hz to 1.5 kHz via wireless BM, measuring over −48 dB attenuated in the other frequencies. Less than ±2% error by inputting a sine wave at pass-band frequencies for FIR algorithm test was obtained between simulated and measured FIR results. Because of the powerful digital FIR design, the total dimension could be dramatically reduced to 23×27×4 mm including wireless BM except for battery. The power isolation was built to avoid the effect of radio-frequency interference on the system as well as to protect brain cells from system damage due to excessive power dissipation or external electric leakage. In vivo performance was evaluated in terms of long-term stability and FIR algorithm for 4 months after implantation. Four TiWiNets were implanted into experimental animals' brains, and single neural signals were recorded and analyzed in real time successfully except for one due to silicon- coated problem. They could control remote target machine by classify neural spike trains based on decoding technology. Thus, we concluded that our study could fulfill in vivo needs to study various single neuron-movement relationships in diverse fields of BMI.
Keywords: brain-machine interface (BMI), wireless neural signal transmission, neural prosthesis, neurodevice